From Cairo, Egypt:
We have purchased a new meter which gets the blood sample from the forearm not the fingers. We have been using another meter for three years now. We use both systems simultaneously, however, the new meter tends to give us readings that are 25-35% higher than our other meter. Would please advise if such discrepancy is expected; if so which one is correct? How should deal with such a problem especially since we use both meters?
My guess is that your older meter it is whole-blood calibrated. The new meter is plasma-calibrated, and a plasma-calibrated meter will report a reading about 8 to 12 percent higher (or perhaps even more) than a whole-blood calibrated meter.
Original posting 5 Jul 2000
Posted to Blood Tests and Insulin Injections
Last Updated: Tuesday April 06, 2010 15:09:11
This Internet site provides information of a general nature and is designed for educational purposes only. If you have any concerns about your own health or the health of your child, you should always consult with a physician or other health care professional.
This site is published by Children With Diabetes, Inc, which is responsible for its contents.
© Children with Diabetes, Inc. 1995-2013. Comments and Feedback.